Model  HDG6202B  HDG6162B  HDG6112B  HDG6082B 
Main Features 
Channels  2  2  2  2 
Waveform Length  64M 
Frequency Range  200MHz  160MHz  110MHz  80MHz 
Sample Rate  1.25GSa/s 
Voltage Resolution  16bits 
Digital Output Channels  16 
Waveforms 
Standard Waveform  sine, square, triangle, pulse, noise, harmonic 
Arbitrary Waveforms  More than 150 types: index rise, exponential decline, ECG signal, Gaussian, semipositive, Lorentz, dualtone multifrequency, DC voltage, etc. 
Frequency Range 
Sine  1uHz~200MHz  1uHz~160MHz  1uHz~110MHz  1uHz~80MHz 
Square  1uHz~60MHz  1uHz~50MHz  1uHz~40MHz  1uHz~35MHz 
Pulse  1uHz~50MHz  1uHz~40MHz  1uHz~25MHz  1uHz~20MHz 
Triangle  1uHz~5MHz  1uHz~4MHz  1uHz~3MHz  1uHz~2MHz 
White Noise  120MHz  120MHz  110MHz  80MHz 
Harmonic  1uHz~100MHz  1uHz~80MHz  1uHz~55MHz  1uHz~40MHz 
Arbitrary  1uHz~50MHz  1uHz~40MHz  1uHz~25MHz  1uHz~15MHz 
Resolution  1uHz 
Accuracy  ±2ppm, 18~28℃ 
Sine Spectrum Purity 
Harmonic Distortion  Typical (0dBm) 
DC1MHz: <60dBc 
1MHz10MHz: <55dBc 
10MHz100MHz: <50dBc 
100MHz160MHz: <40dBc 
THD (Total Harmonic Distortion)  <0.1% (10Hz20kHz, 0dBm) 
Spurious Signal (Nonharmonic)  Typical (0dBm) 
≤10MHz: <65dBc 
>10MHz: <65dBc+6dB/octave 
Phase Noise  Typical (0dBm,10KHz offset) 
0MHz: ≤115dBc/Hz 
Square Wave Characteristics 
Rising/Falling Time  Typical (1Vpp)  Typical (1Vpp)  Typical (1Vpp) 
<8ns  <10ns  <12ns 
Overshoot  Typical (100KHz, 1Vpp) 
<3% 
Duty Cycle  ≤10MHz: 20.0%~80.0% 
10MHz~40MHz: 40.0%~60.0% 
>40MHz: 50.0% (fixed) 
Asymmetry  1% +5ns of Period 
Jitter  Typical (1MHz,1Vpp, 50Ω) 
≤ 5MHz: 2ppm+500ps 
> 5MHz: 500ps 
Trianglular Wave Characteristics 
Linear  ≤1% (1KHz, 1Vpp) of Peak Output 
Symmetry  0%~100% 
Pulse Characteristics 
Period  25ns~1Ms  25ns~1Ms  40ns~1Ms  50ns~1Ms 
Pulse  ≥10ns  ≥10ns  ≥12ns  ≥15ns 
Rising/Falling Time  ≥5ns  ≥6ns  ≥8ns  ≥10ns 
Overshoot  <3% (1Vpp) 
Jitter  Typical (1MHz, 1Vpp, 50Ω) 
≤ 5MHz 2ppm+500ps 
> 5MHz 500ps 
Arbitrary Wave Characteristics 
Waveform Length  64M 
Vertical Resolution  16 Bit 
Sampling Rate  1.25GSa/s 
Rising/Falling Time  Typical (1Vpp): <6ns 
Jitter  Typical (1MHz, 1Vpp, 50Ω) 
≤ 5MHz 2ppm+500ps 
> 5MHz 500ps 
Harmonic Output Characteristics 
Harmonic Times  ≤16 times 
Harmonic Type  Even harmonics, odd harmonics, sequential harmonics 
Harmonic Amplitude  Each harmonic amplitude can be set 
Harmonic Phase  Each harmonic phase can be set 
Amplitude Characteristics ( 50ΩTermination) 
Range  ≤20MHz: 1mVpp ~ 10Vpp 
≤80MHz: 1mVpp ~ 5Vpp 
≤110MHz:1mVpp ~ 2.5Vpp 
≤160MHz:1mVpp ~ 1Vpp 
≤200MHz:1mVpp ~ 0.5Vpp 
Accuracy  1KHz Sine, 0V offset (±1%±2mVpp of setting value) 
Amplitude flatness (relative to 1 kHz sine wave, 500 mVpp, 50 Ω)  ≤1MHz: ±0.1dB  ≤1MHz: ±0.1dB  ≤1MHz: ±0.1dB  ≤1MHz: ±0.1dB 
≤60MHz: ±0.2dB  ≤60MHz: ±0.2dB  ≤60MHz: ±0.2dB  ≤60MHz: ±0.2dB 
≤100MHz: ±0.4dB  ≤100MHz: ±0.4dB  ≤100MHz: ±0.4dB 

≤160MHz: ±0.8dB  ≤160MHz: ±0.8dB 


≤200MHz: ±1.2dB 



Unit  Vpp, mVpp, Vrms 
Resolution  1mV 
Impedance  50Ω 
Offset Characteristics (50Ω termination) 
Range  Voltset < Vmax – Vpp/2 
Accuracy  ± (1% of setting + 5mV + 0.5% of amplitude) 
± ( 1% + 5mV of setting value + 0.5% of amplitude) 
Modulation Types 
Modulation Type  AM, FM, PM, 2ASK, 2FSK, 2PSK, BPSK, PWM 
Amplitude Modulation (AM) 
Carrier Wave  Sine, Square, Triangle, Pulse, Harmonic, Arbitrary (except DC) 
Modulation Source  Internal, external, other channels 
Modulation Wave  Sine, Square, Triangle, White Noise, Arbitrary 
Modulation Frequency  2mHz~50KHz 
Modulation Depth  0%~120% 
Frequency Modulation (FM) 
Carrier Wave  Sine, Square, Triangle, Pulse, Harmonic, Arbitrary (except DC) 
Modulation Source  Internal, external, other channels 
Modulation Wave  Sine, Square, Triangle, White Noise, Arbitrary 
Modulation Frequency  2mHz~50KHz 
PM 
Carrier Wave  Sine, Square, Triangle, Pulse, Harmonic, Arbitrary (except DC) 
Modulation Source  Internal, external, other channels 
Modulation Wave  Sine, Square, Triangle, White Noise, Arbitrary 
Modulation Frequency  2mHz~50KHz 
Phase Deviation  0° to 360° 
2ASK 
Carrier Wave  Sine, Square, Triangle, Pulse, Harmonic, Arbitrary (except DC) 
Modulation Source  Internal, external 
Modulation Wave  50% duty cycle square wave 
Modulation Frequency  2mHz~1MHz 
2FSK 
Carrier Wave  Sine, Square, Triangle, Pulse, Harmonic, Arbitrary (except DC) 
Modulation Source  Internal, external 
Modulation Wave  50% duty cycle square wave 
Modulation Frequency  2mHz~1MHz 
2PSK 
Carrier Wave  Sine, Square, Triangle, Pulse, Harmonic, Arbitrary (except DC) 
Modulation Source  Internal, external 
Modulation Wave  50% duty cycle square wave 
Modulation Frequency  2mHz~1MHz 
BPSK 
Carrier Wave  Sine, Square, Triangle, Pulse, Harmonic, Arbitrary (except DC) 
Modulation Source  Internal 
Modulation Wave  01 yard 
Modulation Frequency  2mHz~1MHz 
Pulse Width Modulation (PWM) 
Carrier Wave  Square 
Modulation Source  Internal, external, other channels 
Modulation Wave  Sine, square, sawtooth, noise, arbitrary 
Modulation Frequency  2mHz~50KHz 
Width Deviation  0% to 100% of Pulse Width 
0% to 100% of the pulse width 
External Modulation Input 
Max. Input Range  75mVRMS to ±2.5Vac+dc 
Input Bandwidth  10MHz 
Input Impedance  1KΩ 
Sweep Characteristics 
Carrier Wave  Sine, Square, Triangle, Pulse, Harmonic, Arbitrary (except DC) 
Type  Linear 
Direction  Top 
Sweep  1ms to 50Ks 
Hold/return Time  1ms to 50Ks 
Trigger Source  Internal, external, manual 
Mark  Falling edge of the sync signal (programmable) 
Burst Characteristics 
Carrier Wave  Sine, Square, Triangle, Pulse, Harmonic, Arbitrary (except DC) 
Carrier Frequency  2mHz to 100MHz  2mHz to 100MHz  2mHz to 80MHz 
Pulse Count  1 to 2000 000 000 
Start/Stop Phase  0° to 360° 
Internal Cycle  2μs to 500s 
Gating Source  External Trigger 
Trigger Source  Internal, external, manual 
Frequency Counter 
Measurement Function  Frequency, period, positive/negative pulse width, duty cycle 
Frequency Resolution  7 bits/s 
Frequency Range  1uHz~200MHz 
Input Level  TTL level 
Gate Time  10ms~16s 
Voltage Range and Sensitivity (Nonmodulated Signal) 
DC Coupling  DC Offset Range  ±1.5VDC 
1μHz to 100MHz  50mVRMS to ±2.5Vac+dc 
100MHz to 200MHz  100mVRMS to ±2.5Vac+dc 
Pulse Width and Duty Cycle Measurement 
Frequency and Amplitude Range  1μHz to 25MHz  50mVRMS to ±2.5Vac+dc 
Pulse Width  Min. Pulse Width  ≥100ns 
Pulse Width Resolution  8ns 
Duty Cycle  Measuring range (display)  0% to 100% 
Input Characteristics 
Input Signal Range  Destruction Voltage  ±5Vac+dc  Input Impedance =500Ω 
Input Trigger  Trigger Level Range  2.5V to +2.5V 
Trigger Sensitivity Range  0% (140mV hysteresis voltage) to 100% (2mV hysteresis voltage) 
Trigger characteristics 
Trigger Input 
Level  TTLcompatible 
Slope  Rise or fall (optional) 
Pulse Width  >50ns 
Reference Clock 
External Reference Input 
Lock Range  10MHz±50Hz 
Level  2.5Vpp to 5Vpp 
Lock Time  <2s 
Input Impedance  5kΩ, AC Coupling 
Internal Reference Input 
Frequency  10MHz ± 50Hz 
Level  3.3Vpp 
Output Impedance  5kΩ, AC Coupling 
Synchronous Output 
Level  TTLcompatible 
Impedance  50Ω, nominal value 
General Characteristics 
Interface  HDG6000B: USB Host, USB Device, Optional RS232 port 
HDG6000C: USB Host, USB Device, LAN port, WiFi, Touch Screen, Optional RS232 port 
Display  7 inch, 64K color, TFT LCD Screen, 800*640 
Voltage  100240V, 45Hz  440Hz 
Power  <50W 
Fuse 

Environment 
Temperature Range  Operation: 10 ° C to 40 ° C Nonoperation: 20 ° C to 60 ° C 
Cooling Method  Fan forced cooling 
Humidity Range  Less than 35 ° C: ≤ 90% relative humidity 35 ° C to 40 ° C: ≤ 60% relative humidity 
Altitude  Operation: 3000 meters or less; Nonoperation: 15000 meters or less 
Mechanical Specifications 
Dimension  318 x 110 x 150mm (L x W x H) 
Weight  3KG 